These radiometers were developed for NIST, to be used with a broadband spectrometer to act as a spectral transfer standard when calibrating other detectors in the 0.25 to 15 μm range.
Each head is composed of a low noise detector, thermistor, TE cooler and heatsink to compensate for any temperature change
Temperature control down to 0.05°C from 20 to 30°C gives a temperature coefficient <0.01 %, thus a voltage output stable to 0.01 %
Plug your TP sensor into the STEP-Controller for power supply and temperature control. You can then use the analog output with a scope or lock-in
Power supply and temperature control for TP series detectors.
There is no user manual available for this product.
You might need these specific products to be able to use the placeholder only.
No compatible product