Ключевые особенности семейства продуктов
Measure, Track and Align
Follow your laser beam wherever it goes.
Unique quadrant detector technology senses laser beam position with high resolution.
For CW, Pulsed and High Rep Rate Lasers
- QUAD-E: energy per pulse from μJ to mJ
- QUAD-P: powers from μW to mW
From UV to FIR and THz
Absorbers to cover all sources, from UV to millimeter wavelengths
Large Area Sensors
9 mm and 20 mm square detectors
Fast USB 2.0 Connection
Ensures full speed tracking
Includes Application Software
Complete LabView application software included, with many features
Совместимые экраны & интерфейсы ПК
Laser position monitor (software included).
Спектральный диапазон0,1 - 3000 μm
Типичное время нарастания150 µs
Типичная чувствительность по мощности2000 V/W
Максимальная частота повторения1000 Hz
Минимальный размер луча4.5 mm Ø
Минимальное позиционное разрешение1 μm
Максимальная измеряемая энергия20 mJ
Эквивалентная энергия шума0,5 μJ
Максимальная ширина импульса2,5 μs
Максимальная средняя плотность мощности0,1 W/cm²
Максимальная плотность энергии50 mJ/cm²
Ширина диафрагмы9 mm
Высота диафрагмы9 mm
Размеры63.5Ø X 40.6D mm
The QUAD-4TRACK is a laser position sensing system designed to support our unique pyroelectric quadrant detectors, QUAD-P and QUAD-E. It is a 4-channel microprocessor-based system that measures the voltage output of each QUAD element and does the math necessary to provide a measurement of the X and Y displacement of a laser beam or image. It is fast and can be used to track, align and/or measure movement in real time, with a resolution of just a few microns!
Our large area pyroelectric quadrant detectors provide unique advantages over other position sensing detectors like silicon quads or lateral effect photodiodes. They are fast, handle high peak power of pulsed lasers without saturation and respond to lasers across the spectrum, from UV to Far IR and even THz. The QUAD-E is intended for use with pulsed sources at up to 1000 Hz, while the QUAD-P is designed for CW and high repetition rate (quasi CW) sources. Both types of detectors can also be used as standalone units, in an analog mode, for incorporation into your own system application. We can provide a lemo pigtail cable for this purpose.
The analog output of the QUAD-4TRACK provides voltage that is directly proportional to the pulse energy or laser power irradiating each QUAD element. When the four voltage outputs are equal, the beam is centered on the QUAD detector. This provides a very useful tool when setting up our QUAD probes with your source for optical alignment.
QUAD-4TRACK includes powerful, stand alone, LabView software which is used to control the instrument, process the data, and display X and Y position. It also displays the energy or power of your source and repetition rate. The large graphic in this screen shows the position of the centroid of the beam and tracks its movement in real time. The software includes many handy features like: set boundary, zoom (2X to 128X), set resolution, data logging, and many more. The green line represents the tracking history.
TRACKING THE BEAM OVER TIME
In the measurement screen shown on the left, we are tracking the beam stability of a pulsed Nd:YLF laser at 10 Hz. The resolution was set at 0.001 nm, the boundary is at 20 µm (red circle), and the zoom feature is at 64X. The total energy is 108.5 µJ, the final position of the laser is at -8 µm in X and -8 µm in Y. The green tracking line shows the movement of the laser about the zero position over a few hundred pulses.
POSITION CALIBRATION SCREEN
We've developed a unique position calibration routine which allows you to calibrate our QUAD-4TRACK system when working with a uniformly round laser beam. It requires the use of a micrometer-driven linear stage (1-axis only). As you can see from the calibration screen on the left, the procedure involves zeroing the instrument, moving the QUAD probe to nine discrete positions (+2.000 to - 2.000 mm) and then capturing the QUAD readings. It then determines correction coefficients (last column) and applies them to the raw data to arrive at "corrected positions". The QUAD probe is now calibrated!