Relative Measurements from 0.1 to 30 THz
Broadband, room temperature operation, easier to use and less expensive than a Golay cell
Wide Dynamic Range from nW to mW
With state of the art pyroelectric sensors, measure down to 50 nW with 1 nW resolution.
Use with T-Rad THz Module or T-Rad-Analog Analog Power Module
Each head can be connected to an oscilloscope using the analog power module (T-RAD-ANALOG) or directly to a PC with the digital power module (T-RAD).
Several Sensor Sizes Available
Choice of 5 mm and 9 mm diameter
Calibrated at 0.63 µm
THZ-B detectors are calibrated at a single wavelength (0.63 µm) and include typical wavelength correction data from 0.25 to 440 µm. They are used for relative measurements outside that range.
SDC-500 Optical Chopper
All THZ-B detectors require the use of an optical chopper, like our SDC-500, to sync the signal at either 5 Hz (DA models) or 25 Hz (DZ models).
User-Friendly Software with Many Features (When used with T-Rad-USB)
- Strip chart, statistics and tuning needle displays
- Controls and power display always visible
- Tuning needle display includes min/max indicators
- Data logging for long term testing
Maximum average power43 μW
Noise equivalent power1 nW
Spectral range110 - 3000 μm
Frequency0.1 - 30 THz
Typical rise time0.2 sec
Typical power sensitivity70 kV/W
Chopping frequency25 Hz
Minimum repetition rate1000 Hz
- 1. Projected spectral range. From 10 to 440 μm, spectrometer measurement. From 440 to 3000 μm, relative measurement only. This spectral range is subject to change.
- 2. SDC-500 digital optical chopper sold separately
Maximum average power density10 MW/cm²
Aperture diameter5 mm
Dimensions66.0Ø x 46.5D mm
How to set up and use our THZ-B-DA and THZ-I-BNC analog radiometer probes
THZ-WC-13 - Winston cone accessory for THZ-B
How calibration works
Understanding the spectral characteristics of your detectors
A breakthrough in THz calibration
There is no document to download for this product. Contact us for technical details.
The T-RAD is a microprocessor-based digital radiometer that includes a 12-bit ADC and unique DSP lock-in software. It is powered by a USB connection, which also acts as a virtual COM port. When a THZ-B terahertz pyroelectric detector is plugged into the T-RAD module, the module reads the content of the head's EEPROM, which identifies the detector and provides calibration and wavelength correction data. The LabView software supplied with the device makes it very easy to set up the radiometer, measure a THz or broadband source and record data. The software is compatible with Windows XP, VISTA and 7.
INSTRUMENT CONTROL AND STRIP CHART
Instrument controls and the radiant power measurement are always visible, making it easy to change the radiometer settings, no matter which display tab is selected. Instrument controls include: range, filter tau, batch size, data collection mode, reset options, and a null button for background cancellation. In addition, there are more set up and operation status indicators including: detector Rv, wavelength, frequency (actual), locked and frequency in range lights. The strip chart displays the radiant power measurement in watts, either continuously or by the batch. Select full scale, auto scale or use our manual scaling option.
INSTRUMENT CONTROL AND TUNING NEEDLE
The "tune" tab selects the very useful "tuning needle" display. This is a simulated analog meter whose speed is determined by the "filter tau" setting. It is expected to be used during the set-up of a radiometer with a source. The "tau" value is usually set to a small value when aligning the probe to the source (i.e. when peaking the reading). There is a button control to select "full scale", "min-max" or "reset". In the "min-max" mode, the indicators are "blue" for the minimum power and "red" for the maximum power.
INSTRUMENT CONTROL AND STATISTICS
In the "Statistics" tab there are 4 large windows that contain the statistics for the selected batch, including: Minimum, Maximum, Standard Deviation and Mean, expressed in Engineering Notation. Standard Deviation can be displayed in Watts or as a % that is user-selectable. There is also a window that shows the bandwidth of the Digital Band Pass Filter based on the user selected "Filter Tau" (0.100 to 100 seconds). A lower time constant is helpful when setting up, and a longer one when making measurements, especially on the lower ranges of the instrument.
THZ-B-DA DETECTOR AND OSCILLOSCOPE
Here is a basic analog set up that would be useful if the optical power of the source was about 5 µW or greater. The output of the THZ5B-BL-DA detector would be approximately 600 mV @ 5 Hz chopping frequency, giving plenty of signal for an oscilloscope. Simply read the voltage output and divide by the Rv factor (V/W) of the detector to measure the intensity of the source in Watts. Also consider applying a wavelength correction factor under certain circumstances.
THZ-B-DA DETECTOR AND LOCK-IN AMPLIFIER
This is another analog set-up option that we recommend if you have to measure very low power levels (i.e. less than 5 ?W) where the signal may be buried in the broadband noise. The voltage output of the analog THZ-B-DA detector, powered by our T-RAD-ANALOG, is routed to the Lock-In Amplifier input, and the Sync Output of our SDC-500 Chopper is connected to the reference input. The Lock-In Amplifier will lock on the chopping frequency and you can dial in a long integrating time and measure a very low RMS voltage. The voltage divided by our Voltage Responsivity (V/W) equals the power of the source.
THZ-B-DZ DETECTOR AND T-Rad MODULE
Although analog solutions are available, for simplicity, convenience and sensitivity, we recommend you choose our THZ-B-DZ detectors and the T-RAD digital radiometer. Our unique DSP Lock-In Amplifier software provides a function much like the Analog Lock-In, but is so much easier to use. It also addresses thermal drift of the sensor and allows you to display the power measurement and complete statistics directly in digital and graphic formats. Set the range, null the background, set the filter tau (bandwidth) and make the measurement. It's that easy!
Compatible products you may need
Your item has been added to your cart!
You might need these specific products to be able to use the placeholder only.
No compatible product