USB 3.0 for the Fastest Transfer Rates
Up to 10X faster than regular USB 2.0 connections (also USB 2.0 compatible)
For accurate profile measurements of very small beams: 5.5 µm pixel pitch for the BEAMAGE-4M
- 11.3 x 11.3 mm for the BEAMAGE-4M and BEAMAGE-4M-IR
- 20.5 x 20.5 mm for the BEAMAGE-4M-FOCUS
Available with IR Coating
The BEAMAGE-4M-IR cameras have a special phosphor coating for IR wavelengths (1495-1595 nm)
D4σ definition of diameter, centroid, ellipticity and orientation are ISO 11146:2004 and 11146:2005 compliant
Intuitive Software Interface
Easy to navigate interface, with many display and control features:
- 2D, 3D and XY displays
- Background subtraction function
- Unique "animate" function
- Gaussian fit
- Semi-log graph
To synchronize the camera with a pulsed laser
Software development kit
A software development kit is available. Learn more about BEAMAGE-SDK.
Spectral range350 - 1150 nm
Pixel multiplication factor (PMF)1.8 typical
Minimum measurable beam120 μm
Sensor technologyCMOS (with Fiber Optic Taper)
Pixel count4.2 MPixels
H x V2048 x 2048
Beam diameter definitionsD4σ (ISO compliant)
1/e2 along crosshairs (13.5%)
FWHM along crosshairs (50%)
Frame rate6.2 fps @ 4.2 MPixels (Full Frame)
ADC level12 bit (default) / 10 bit (option)
RMS noise1000:1 (60 dB)
Maximum power density110 W/cm²
Maximum energy density2100 mJ/cm²
Maximum power31 W
- 1. At 1064 nm. With ND4 filter.
- 2. At 1064 nm, 10 ns. With ND4 filter.
- 3. At 1064 nm. With ND4 filter.
Dimensions61H x 81.1W x 46.5D mm
Pixel dimension110 μm
Active area20.5 X 20.5 mm
- 1. With a typical pixel multiplication factor (PMF) of 1.8.
Simulated beams for BEAMAGE
Software, drivers and firmware
Software, drivers and firmware for BEAMAGE and BEAMAGE-M2.
LabVIEW VI examples (32-bit)
LabVIEW VI examples (32-bit) for BEAMAGE series.
LabVIEW VI examples (64-bit)
LabVIEW VI examples (64-bit) for BEAMAGE series.
.NET framework example
.NET compatible example for integration of BEAMAGE beam profiling cameras through a software pipeline. Note that the PC-BEAMAGE software must also be installed and running.
USB drivers installer
Use this file to install USB drivers for these products: MAESTRO, INTEGRA, U-LINK, P-LINK, M-LINK, S-LINK, BLU, PRONTO, HP, MACH6, BEAMAGE
INTUITIVE SOFTWARE INTERFACE
1 - MAIN CONTROLS
The upper part of the software includes all the main controls in a ribbon format. These controls are grouped by family: capture controls, file controls, buffer controls, M2 controls and data computations. The last includes very useful filters and a normalizing function.
2 - DISPLAYS
The left-hand side of the software is the display panel. Four displays are available: 3D, 2D, XY (cross-sectional graphs along the crosshairs) and beam tracking. The desired display is selected by clicking on the corresponding icon at the bottom of the panel. Print screen controls are available for the 3D, 2D and beam tracking displays. They allow the user to save an image of the current view in BMP format.
3 - HOME AND SETUP TABS
The right-hand side of the software contains the home, setup and data acquisition tabs. The home tab allows the user to select the main controls for his measurements (beam diameter definition, crosshair center and orientation) and displays the resulting measurements below. The setup tab allows the user to set the measurement parameters (exposure time, image orientation and averaging, active area, etc.) and the data acquisition tab allows the user to save measurements or measurements and full images, to enter the sampling rates and a total duration for the acquisition. More tabs with advanced controls are available when clicking on the show/hide options button in the computations panel.
3D, 2D AND XY DISPLAYS
The 3D display shows the actual shape of the beam. It is possible to easily zoom, pan and rotate the image. The very useful reset button allows to put the data back in its original configuration. This display also features a print screen button to save the latest image in BMP format.
The 2D display features the crosshairs (set to the major and minor axis or along specified angles) and the measured diameters of the beam. These diameters vary with the chosen definition (4-sigma, FWHM, 1/e2, etc.) and the display can be turned on or off. The print screen button allows to save a picture of the current screen in BMP format.
The XY display plots cross-sectional graphs of the beam along the crosshairs. This display features many useful tools like zoom, cursor, and FWHM and 1/e2 level bars. It is also possible to display the graphs in semi-log format to enhance the details in the low intensity parts of the beam.
BEAM TRACKING DISPLAY
The beam tracking display allows the user to visualize the variation of the beam's centroid position on the sensor. This display shows the latest calculated position as well as the previous ones, until the user resets the view. The display also shows the mean position of all computed values and gives information regarding position stability for both X and Y axes. This tool is great to monitor the beam pointing stability over time.
Compatible products you may need
Your item has been added to your cart!
You might need these specific products to be able to use the placeholder only.
No compatible product